The LUNA-II™ is an automated cell counter that gives you the ultimate brightfield cell counting experience. Press count and the LUNA-II™ does the rest – bringing your cells into the sharpest focus, adjusting light levels, capturing a high-resolution image, and analyzing the image to produce accurate cell count and viability data. You can create custom, cell-specific cell counting protocols or simply use the default counting protocol suitable for most cell lines. Live cells are tagged with green circles and dead cells are tagged with red circles, making it easy to verify the accuracy of each count. This little counter sits comfortably in a cell culture hood or on your benchtop to fit seamlessly into your workflow.
Just awesome! It is definitely a cost efficient choice!
Working with LUNA-II automated cell counter is so easy and efficient in comparison to counting cells under the microscope. The single-use slides are so easy to use and the LCD display gives you the opportunity to view in real-time what's being counted. Another advantage is the standardised protocols which also save a lot of time and effort in order to optimize your daily work with the apparatus.
Plovdiv Medical University
Great results! Will continue to use it.
Logos Luna Cell Counter system saves us thousands of hours counting cells for each passage. It's easy-to-use, fast, reliable, and overall cost-effective, especially when using the reusable slides.
University of the Ryukyus
Overexpression of MMPs, cytokines, and RANKL/OPG in temporomandibular joint osteoarthritis and their association with joint pain, mouth opening, and bone degeneration: A preliminary report.
2021. Cafferata EA, Monasterio G, Castillo F, Carvajal P, Flores G, Díaz W, Fuentes AD, Vernal R. Oral diseases 27(4):970–980.
Aryl Hydrocarbon Receptor is Involved in the Proinflammatory Cytokine Response to Cadmium.
2021. Kulas J, Tucovic D, Zeljkovic M, Popovic D, Popov Aleksandrov A, Kataranovski M, Mirkov I. Biomedical and Environmental Sciences 34(3):192-202.
Inertial Microfluidics-Based Separation of Microalgae Using a Contraction–Expansion Array Microchannel.
2021. Kim GY, Son J, Han JI, Park JK. Micromachines (Basel) 12(1):97.
Mitochondrial pyruvate carrier 1 regulates ferroptosis in drug-tolerant persister head and neck cancer cells via epithelial-mesenchymal transition.
2021. You JH, Lee J, Roh JL. Cancer Letters 507:40–54.
Interplay Between Nanoplastics and the Immune System of the Mediterranean Sea Urchin Paracentrotus lividus.
2021. Murano C, Bergami E, Liberatori G, Palumbo A, Corsi I. Frontiers in Marine Science https://doi.org/10.3389/fmars.2021.647394.
2022-04-07 | 648.38 KB
2022-04-07 | 2.14 MB
2021-06-18 | 266.55 KB
2018-04-13 | 906 KB
2018-04-16 | 906 KB
2018-04-13 | 665.84 KB
|L40001||LUNA-II™ Automated Cell Counter (with built-in printer)||1 unit|
|L40002||LUNA-II™ Automated Cell Counter (without printer)||1 unit|
|L12001||LUNA™ Cell Counting Slides, 50 Slides||1 boxes|
|L12002||LUNA™ Cell Counting Slides, 500 Slides||10 boxes|
|L12003||LUNA™ Cell Counting Slides, 1000 Slides||20 boxes|
|L12011||LUNA™ Reusable Slide||1 unit|
|L12012||LUNA™ Reusable Slides||2 units|
|L12014||LUNA™ Reusable Slide Coverslips||10 units|
|T13001||Trypan Blue Stain, 0.4%||2 x 1 mL|
|L13002||Erythrosin B Stain||2 x 1 mL|
|B13101||LUNA™ Standard Beads||2 X 1 mL|
|L44003||LUNA-II™ IQ/OQ Protocol||1 copy|
|P12002||LUNA-II™ Printer Paper (Thermal, 830 prints/roll, 10 rolls)||2 x 5 rolls|
|Cell Counting Time||less than 10 s (manual focusing), less than 15 s (autofocusing)|
|Cell Concentration Range||5×104 – 1×107 cells/ml|
|Cell Size Range||3~60 μm (optimal : 8~30 μm)|
|Cell Viability Range||0~100 %|
|Image Resolution||5 MP|
|Image Type||TIF, annotated TIF|
|Dimensions (W×D×H)||16 × 18 × 28 cm (6.3 × 7.0 × 11.0 in)|
|Weight||1.6 kg (3.5 lb) *without the AC adaptor and power cord|
|Operating Power||100~240 VAC, 1.2 A|
|Electrical Input||12 VDC, 3.3 A|