A comparison of yeast cell counting methods
LUNA-II YF™ Automated Yeast Cell Counter vs. flow cytometry vs. hemocytometerdownload pdf
- LUNA-II YF™ Automated Yeast Cell CounterThe LUNA-II YF™ Automated Yeast Cell Counter is a fully automated solution for yeast cell counting and viability analysis that fits on any benchtop. Dual fluorescence optics, an autofocusing liquid lens, and a proven counting algorithm produce cell count and viability data in just 15 seconds, doing away with the subjectivity and time of manual cell counting. The LUNA-II YF™ eliminates the subjectivity of operator judgment and guarantees that cells are counted with the same criteria between counts and users. Data may be transferred to a USB drive or printed on an external printer for effortless recordkeeping. The LUNA-II YF™ also generates comprehensive PDF reports with count data, protocols, corresponding cell images, and relevant histograms.
- Accuracy compared to a flow cytometerTo evaluate the accuracy of the LUNA-II YF™, cell counts were compared to those obtained with the BD FACSCalibur™ flow cytometer (BD Biosciences). Serial dilutions of Sacchromyces cerevisiae were counted in triplicate. For the LUNA-II YF™, yeast cells were labeled with Acridine Orange/Propidium Iodide Stain, loaded into a disposable PhotonSlide™, and counted with the default AO-PI protocol. For the flow cytometer, yeast cells were labeled with a thiazole orange/propidium iodide stain and standard beads were added to help determine cell concentration. There was no significant difference between the LUNA-II YF™ and the flow cytometer in determining cell concentration (Fig 1) or cell viability (Fig 2), demonstrating the accuracy and reliability of the LUNA-II YF™.
- Figure 1. Comparison of the yeast cell concentration results from the LUNA-II YF™ and a flow cytometer.
- Variability compared to a hemocytometer and flow cytometerTo compare the variability of cell concentrations obtained with the LUNA-II YF, flow cytometer, and hemocytometer, serial dilutions were prepared and counted in triplicate. To test the hemocytometer (Marienfeld), cell samples were mixed with an equal volume of 0.4% trypan blue stain. The cells were imaged with the iRiS™ Digital Cell Imaging System and the cells within the four corner squares of the Neubauer counting grids were counted. There is no significant difference in the total concentrations between any of the methods (Fig 3). However, the hemocytometer showed a higher variability from count to count whereas the LUNA-II YF™ and the flow cytometer were more consistent.
- Figure 2. Comparison of the yeast viability results from the LUNA-II YF™ and flow cytometer. (A) Viability results measured by the flow cytometer. (B) There was no significant difference in the viability results obtained with the two instruments.
- SummaryThe LUNA-II YF™ Automated Yeast Cell Counter is a cost-efficient and simple alternative to expensive flow cytometers as well as being faster and more accurate than manual cell counting.
– cell count and viability data in just 15 seconds with a few taps on the touchscreen
– no user-to-user variability
– a tried, tested, and trusted cell counting algorithm
– automatically generated data reports with cell images, histograms, protocols, and data
- Figure 3. Comparison of the variability of results from a flow cytometer, the LUNA-II YF™, and a hemocytometer. The variability of results is greater for the hemocytometer than for the other two instruments.
- Using the LUNA Reusable Slide for accurate cell counting with automated cell counters
- How to Count Cells An Overview of Cell Counting Methods
- Fast automated yeast cell counting algorithm using bright-field and fluorescence microscopic images
- LUNA-II YF™
- Logos Biosystems Products